Terak 8510/c Info?
derschjo at mail.msu.edu
Sat Nov 6 14:00:30 CDT 2010
On 11/6/2010 9:11 AM, Al Kossow wrote:
> On 11/6/10 2:11 AM, Josh Dersch wrote:
> main difference I see is the keyboard and monitor connector panel.
> It's probably a bit of a hassle, but could you get a picture of the
> guts of the keyboard/montor back panel assembly? The keyboard looks
> like a Keytronics with a serial interface, similar to the ones we
> used on the later AED terminals.
> I'm guessing they added a deserializer to the back panel assembly.
Sure thing. I'm out of town this weekend, but I'll take some pictures
early next week. (I need to take the thing apart anyway to get all the
bits of old foam out of it.)
The good news is the power supply seems to be OK (the voltages are
actually listed on a panel on the inside I didn't see the first time I
went over it.) The bad news is it doesn't do much when I power it on.
It blinks the drive access light in a regular pattern (I'm wagering this
is a diagnostic code) but that's it. I don't suppose anyone knows what
these codes are?
More information about the cctech