olivetti programma 101 repair
simski at dds.nl
Wed Dec 17 05:09:23 CST 2014
Good news! I've located the problems with the machine and already have
fixed the mechanical part. The elctrical problem is bad contacts in the
edge connectors (a mere 850 pins...) and I got my first calculating
results on paper in the form of the square root of 6. :-)
So ignore my earlier post on the scope probe path
updates of course on the blog.
On 14-12-14 18:25, Simon Claessen wrote:
> hello all,
> I'm in the process of repairing our 101 but i hit a solid wall finding
> out whats wrong with it. are there people with a working copy who could
> do some measurements on their copy?
> Unfortunately i have no schematic of the electronics, only the mechanics
> and PSU.
> I have made a crappy video of the scope trace of the delay line input
> and i'm very interested in a video of a working machine's delay line. :-)
> The video is here: http://youtu.be/llpwjGeldt0
> and a partially repairlog is here:
> als scematics are welcome as well. :-)
Met vriendelijke Groet,
More information about the cctech